At 100 MHz (later slightly down-rated) with the Type 80 plug-in and up to 85 MHz with the dual-channel Type 82, the 585 was the highest-speed general purpose scope of Tek's tube scope era (the 1 GHz 519 was specialized with no vertical amplifier).
The 585's front panel layout is almost identical to the 545A that came out at the same time. The 585 also shares internal construction and circuits such as the power supply with the 535/545 line of scopes.
|Rise time||3.5 ns with Type 80 plugin and P80 probe|
|Timebase A||50 ns/Div to 2 s/Div, 1−2−5, variable ×2.5, magnifier ×5, single sweep; X input|
|Timebase B||2 μs/Div to 1 s/Div, 1−2−5, can delay timebase A by 1 μs to 10 s|
|CRT||T581, distributed vertical deflection plates, 10 kV acceleration|
|Outputs||A Sawtooth, A Gate, B Gate, Calibrator (0.2 mV to 100 Vp-p, 1−2−5)|
The 585A uses a tunnel-diode trigger circuit (see below), adds HF Sync and AC LF reject modes to the timebase A trigger selector, and adds a trace rotation coil.
The 585 uses 580-series plug-ins which look like letter-series and 1-series plug-ins but are electrically different. The main difference is the impedance of the signal that passes through the interface from the plug-in to the mainframe. In 530/540/550-series scopes, the plug-in expects the mainframe to be a high impedance load. Indeed, in those scopes, the plug-in is directly driving tube grids in the mainframe. In 580-series scopes, the plug-in expects the mainframe to have a controlled impedance of 93 Ω per side (differential signal). The pin-out of the interface connector is also different, the 580-series pin-out being essentially flipped 180° from its 530/540/550 series counterparts (Signal Out is between pins 14 and 16 as opposed to between pins 1 and 3, etc.)
The Type 81 adapter allows the use of letter-series and 1-series plug-ins in a 585.
The original 585 (up to serial number 1070) has an "A" trigger circuit that has a differential trigger amplifier made of a pair of 6EW6 miniature pentodes, followed by a Schmitt trigger made of a pair of miniature dual triode tubes.
Starting at serial number 1071, the miniature dual triode Schmitt trigger was replaced by a 10 mA tunnel diode and a pulse amplifier made with an OC171 Germanium PNP transistor. The tunnel diode is AC-coupled to the base-emitter junction of the transistor.
Vertical Signal Path
Like the 545, 551, and 555, the 585 uses two distributed amplifiers in the vertical signal path, one with seven differential 6DJ8 stages driving a helically-wound, differential delay line, followed by a five-stage differential distributed driver amplifier and a 7699 power stage driving the CRT.
Unlike the other 500-series scopes, the 585 uses a CRT with distributed vertical deflection plates, the T581, enabling it to have both high sensitivity and fast risetime. Total acceleration voltage is 10 kV.
Several improvements were published for the 585 circuitry, including the enhanced trigger circuit. If all of those changes are applied to a 585, it is essentially a 585A.
The 585 power supply is similar to other Tek scopes of the era such as the 549. A linear regulator with a 5651 reference tube, a 12AX7/6AU6 error amplifier and three parallel 12B4 series pass tubes produces −150 V. This is the only adjustable rail, all other regulated voltages are referenced to it. The +100 V and +350 V regulators use half a 6080 each as the pass element, the +225 V two. In the +500 V path, a 12B4 performs this function. The unregulated +325 V input to the 225 V supply is used to supply the HV oscillator, a 6AU5.
A transistor-based regulator supplies a +12.6 V heater voltage to the plug-in.
A 6N045 delay relay controls a multi-contact relay to switch supply voltages on only after the tubes are warmed up. When the main relay engages, the delay relay is cut out so it will be ready to delay again even after a quick interrupt in power input.
The 585 contains a 53°C (128°F) thermal cutoff.
The 585 uses semiconductor diodes as rectifiers and Germanium transistors in the 12 V power supply.
Rackmount versions were also made, the RM585 and RM585A.
585 with Type 82