The Tektronix Type 507 was designed for the power industry for surge testing. It was introduced in 1959.
The vertical signal path of the 507 does not contain amplifiers. It has a 5 nanosecond risetime. The input impedance of the 507 is 72 Ω. The maximum sensitivity is deliberately low, at 50 V/cm.
The input attenuator is very unusual for Tek scopes. It is a switch with ten positions in "% of signal", where each step is 10%. The deflection signal is permanently AC-coupled and, unusually, the DC blocking capacitor is after the input attenuator.
The 507 has a total acceleration voltage of 24 kV, produced by four 1X2 tubes. The lower-voltage power supplies, located in the external box, are all tube except for selenium rectifiers for some of the lower voltages.
The 507 uses the T507 CRT.
- Service Scope No. 18, Feb 1963: Anode-Connector Arcing in the Type 507 Oscilloscope
507 with 202 cart