If you get the "Sorry, the file cannot be displayed. There seems to be a technical issue ..." error when trying to view an image, a workaround is to click on the cog icon in the upper right side of the media viewer, and then select "Disable Media Viewer". We are aware of the problem and are looking for a solution.
The Tektronix 2201 is a combined 10 MHz / 20 MS/s analog/digital oscilloscope introduced in 1988.
|Rise time||17.5 ns|
|Bandwidth||20 MHz (analog)|
|Deflection||5 mV/Div to 5 V/Div, 1-2-5, ±3%|
|Sweep (analog)||50 ns/Div to 0.5 s/Div, ×10 magnifier (5 ns/Div)|
|Sweep (digital)||20 μs/Div to 0.5 s/Div, ×100 store to 50 s/Div)|
|Input impedance||1 MΩ // 25 pF|
|Sampling rate||10 MS/s, both channels simultaneously (two ADCs)|
|Record length||2K per channel|
|Resolution||vertical 8 bit, horizontal 11 bit|
|Probes||Two P6103 10X 2M Probes|
|CRT||8 cm × 10 cm, 12.5 kV acceleration|
|Power||95−128 VAC / 185−250 VAC, 48−440 Hz, 70 W|
|Size and weight||138 mm (w) × 380 mm (h) × 438 mm (d), 7.6 kg|